The calibration process for the oxygen sensor for my Nitrogen Sensor is complicated and time consuming.
Pneumatech use advanced Zirconia Sensors on all its PSA models. Therefore there is no need to calibrate the sensor at the site itself. Zirconia sensors are the most reliable sensors available in the market for tracing oxygen content in the gas streams.
For the membrane units:
Pneumatech's new PMNG 1-3 Series has an user-friendly calibration button located on the front of the machine for a quick and easy calibration process. The nitrogen analyzer is also battery-powered, and will shut down after a few minutes of inactivity to preserve battery. To perform the nitrogen analyzer calibration on your PMNG 1-3, follow the simple steps below:
Turn switch to "calibrate"
Check pressure regular setting at 7.25 --> 10.15 psi
Press "On", then press "Cal"
Let unit display stabilize at 79.1%
Turn switch back to "Monitor N2"
To browse Pneumatech's line of nitrogen generators, please click HERE. Have a question about our nitrogen generators? Email firstname.lastname@example.org.